"I was reading one of your old posts about magnets and I was wondering if a magnetic front closure on a bra would be a problem? There's a warning on the label but I know part of that is just due to liability. What about the new Victoria Secret bra that has a magnet clasp on the front? If the magnet hits right in between the breasts would it be close enough to the device that it could interfere? Also does having a magnet that close change the settings or turn off a defibrillator/pacemaker early? I'm sure most doctors would say just wear another bra but this bra in particular is very comfy! I've tried it on but not worn it for extended periods of time. Luckily this is one of the only major complaints I've had about having heart disease and a medtronic device at such a young age."First, let me say thank you for asking this question: who knew research could be so, er, entertaining! Second, this question reinforces why medical blogging is so great: you learn something new every day.
Now, as I slap myself back to a bit more professional stance, I'll summarize by saying I think you'll be okay to use such a bra with some precautions. Given the picture and the clasp's location, this bra is more likely to interfere with the pacemaker of the partner you hug rather than yourself, provided your pacemaker was implanted over 3 centimeters from the magnetic clasp. Since most pacemakers and defibrillators are implanted just below the collar bone, the chance of the magenetic clasp to interfere with your device is remote. From my earlier post:
Pacemakers and defibrillators contain a small "reed switch" that is sensitive to magnetic fields that allows patients and their doctors an opportunity to affect their device to perform specific functions outlined below.-Wes
In the case of pacemakers, the activated reed switch tells the pacemaker to pace, irrespective of the person's underlying rhythm, at a specific rate (determined by the manufacturer of the pacemaker). When the rate changes, this tells doctors how much voltage is left in the person's pacemaker, and uses the paced rate to act like a battery meter, telling doctors when the pacemaker battery voltage is getting low. It does NOT inhibit pacemaker output. (Oh, there will be some wise guy that says that pacing that does not synchronize with one's heart rhythm could land in the "vulnerable period" of the cardiac cycle and induce an abnormal rhythm (and yes, that can occur), but magnet checks are done tens of thousands of times a day in the US and I have never heard of someone dying from this with conventional pacemakers).
In the case of a defibrillator (that treats abnormally fast and slow heart rhythms), the reed switch acts slightly differently. Again, a magnet over the person's defibrillator does NOT inhibit pacing at all. In the case of a defibrillator, a magnet over the device that is powerful enough to trip its reed switch will suspend detection of rapid heart rhythms while the magnet is over the device. In the case of this article, this will only happen if a magnet is held within three centimeters of the device. That's only 1.5 inches, folks. In other words, one of these fridge magnets or pieces of jewelry would have to be held virtually right over the device to have any effect. Certainly, if the person had the unfortunate luck that a rapid heart rhythm occurs when a magnet of sufficient strength is over the device, then the device would not detect this rapid heart rhythm and it could be fatal. But the odds of that happening are very, very low.
Wolber T, Ryf S, Binggeli C, Holzmeister J, Brunckhorst C, Luechinger R, Duru F. Potential interference of small neodymium magnets with cardiac pacemakers and implantable cardioverter-defibrillators. Heart Rhythm. 2007 Jan;4(1):1-4. Epub 2006 Sep 16.